Posted - 02/24/2017 : 04:36:24
I have had my 300 for several years.. It used to work fine.. but has been sitting in a drwaer for years with no battery.
Recently I decided to se what was happening with it, and installed a new battery. After it came to life I nticd the right hand side of the display incl the battery level meter display is faded..but more important is zero clicks.
I dont have a test source, so cannot check or calibrate.
I tried testing the tube voltage, but the only meeter I have is an el cheapo yellow china digital version with max 500V DC range.
On that setting i get zero v on the tube??
Is there a circuit for this item, or are there some test points on the back of the board I can check.
While testing.. I noted it will count clicks when I touch the Multimeter probes.. I can make it count up to the alarm level of 100 and the alarm goes off. This actually shows the device is working.. but maybe the tube is not. Its had very little use..??
Is it likely its working and I really do have a zero BG radiation??
or is there some problem?
I did a factory reset, but no change.
Hope someone can comment soon , would be good if you can reply to my e mail below.
Gary in Bali
Contact me here