Hi All, The average Counts per minute at my station started going haywire sometime yesterday - was at 17.21 then skyrocketed without a spike in actual CPM data. Now its moving around between 2 and thousands of CPM. Any idea what is going on?
Yeah looks like there's still a but that isn't fixed. It has been reported before but it's hard to produce because it happens every 45 days that the ACPM just spikes. There's no way to fix the ACPM unless you restart the unit.
Looks like the problem is in the number of total minutes. 60*24*45 = ~65k. The GMC300/320 units are only 16 bits an can't handle those number I think so it's going to roll back to 0. We will check the code.
Hello, a new firmware has been released for a fix on this issue. If there's still a problem after 45 days please let me know. Also, please email me with your serial number for a firmware upgrade. Thanks.